Prime Products Ltd
90 D. Moutsopoulou Str.
Piraeus 185 41, Greece
+30 210 4819800
+30 210 4203523
Prime Products Distribution (S) Pte Ltd
48 Toh Guan Road East # 02-124
(Enterprise Hub) Singapore 608586
Prime Products Distribution FZC Ltd,
Kuwait Road, Block B7,
3rd Floor, Office 301,
Akron Bldg., P.O. Box 50057 Fujairah UAE
+971 9 2283203
+971 9 2283212
+971 9 2283204
Prime Products Distribution B.V.
33 Pittsburghstraat Str
3047 BL Rotterdam
+31 (0) 10 462 3055
+31 (0) 10 245 9026
Prime Products Distribution USA LLC
6933 Clinton Dr - Houston
TX 77020, USA
+1 713 893 4345
+1 409 888 9424
+1 713 268 3298
P.P.D. Prime Products Distribution Ltd.
5A, Amigdalias, Agia Filaxis
3112, Limassol, Cyprus
How to calibrate the Elcometer 456 Coating Thickness Gauge in accordance with ISO 19840, for rough surfaces (or blast coated substrates), on ferrous substrates.
Get the best performance from the Elcometer 456 Coating Thickness Gauge and make the best use of the gauge’s ±1% accuracy by calibrating the coating thickness instrument to suit the application.
The Elcometer 456 Dry Film Thickness Gauge can measure the thickness of a coating on smooth, and on rough or blast coated substrates; and there are a number of calibration methods for each substrate profile.
When measuring on rough surfaces, if you have access to a sample of the uncoated substrate you will be measuring, then you should undertake a rough surface (or 2-point) calibration.
However, if you don’t have access to the uncoated substrate, then you can use the Elcometer 456’s patented Zero Offset calibration method, which helps measure film thicknesses in accordance with ISO 19840.
ISO 19840 simply states that if a blast profile is unknown, you should calibrate on smooth steel and then subtract a predefined correction value from the measurement taken on the coated surface.
The Elcometer 456’s patented zero offset calibration automatically subtracts this correction value from the measurement, so you don’t have to.
The Elcometer 456 is an intelligent gauge, so, once the gauge has been set to the Zero Offset Calibration, it will guide you through the calibration process, one step at a time.